Genevac EZ-2 Plus Solvent Evaporator Mark 3
MN: EZ2.3 NOT HCL COMPATIBLE
Includes: Rotor, power cords, and vessels as pictured. Buckets not included.
Condition: Used in good working condition. Last serviced Sept 2020. See full condition report below.
- Self-test passed. Vacuum held for 5 hours, rotor spun without error, temp held.
- Software revision: v8.04 (shown at startup)
- System/component hours:
- Pump: 3379
- Lamp: 1055
- Compressor: 3680
- Heater: 108
- Rotor: 3282
- (accessible via bottom dial, maintenance)
- 1000mbar to 10mbar in ~5 minutes @70F ambient
- 1000mbar to 5mbar in ~10 minutes @70F ambient
- holds ~5mbar through testing cycle (5hr)
WORLDWIDE SHIPPING AVAILABLE
30 Day Returns